Sure Status COVID-19Ag CardTest, kit/25T
Sure Status COVID-19 Antigen Card Test, kit/25 tests
Indicative Price 48.75 USD
Sure Status COVID-19 Antigen Card Test is a lateral flow immunochromatographic assay for the qualitative detection of nucleocapsid protein antigen from SARS-CoV-2 in nasopharyngeal (NP) swab specimens directly collected from individuals who are suspected of COVID-19 by their healthcare provider. Sure Status COVID-19 Antigen Card Test is for in vitro diagnostic use and intended as an aid to detection of nucleocapsid protein antigen in patient with clinical symptoms of SARS-CoV-2 infection.
Technology: Lateral flow immunochromatographic test
Format: Cassette, 25 tests
Sample type: Nasopharyngeal swab
Analytical sensitivity: Limit of detection (LOD) in natural nasopharyngeal swab matrix was confirmed as 22.4 TCID50/swab
Analytical specificity: No cross-reactivity or interference was seen with 16 viruses, 13 micro-organisms and 2 negative matrixes
Clinical sensitivity: 94.16%
Clinical specificity: 100%
Manufacturer product reference: Premier Medical Corporation Pvt.
Items supplied with:
·25 x Test device pouch containing: test device and desiccant
·25 x Nasopharyngeal swab
·25 x Reaction buffer vial with nozzle
·3 x Extraction Buffer Bottle (3 ml)
·1 x IFU
Items required, but not supplied:
New pair of disposable gloves & facemask, Permanent marker pen, Timer, PPE Kits and Biohazardous waste container
24 months from the date of manufacturing; do not use beyond its expiry date.
Weight & volume:
Estimated weight: 0.33kg
Estimated volume: 0.00202m3
Storage & transportation condition:
Must be stored at 4 - 30°C. Do not freeze the kit. The kit is sensitive to humidity and heat. Do not store the kit at temperatures above 30°C and in humid conditions.
Packaging & labelling:
Primary packaging: All contents of Sure Status COVID-19 Antigen Card Test, kit of 25 tests
Secondary packaging: One kit Sure Status COVID-19 Antigen Card Test, kit/25 tests
·Read the instructions carefully before performing the test, any deviation will invalidate the test results.
·Do not re-use the test kit.
·Do not use the test kit if the pouch is damaged or the seal is broken.
·Do not use the extraction buffer bottle from different lot.
·Wear personal protective equipment, such as gloves and lab coats when handling kit reagents. Wash hands thoroughly after the tests are done.
|To see icon descriptions, go to |
' Product attribute guide'