Cannula IV short 24G, ster, disp, BOX-50
Cannula, IV short, 24G, sterile, disposable, BOX-50
Indicative Price 6.37 USD
Cannula, IV short, 24G, sterile, disposable
Length expressed in mm.
External diameter expressed in Gauge and mm.
Colour code/external diameter: Visible at the base of cannula.
Components: Protecting cap, trocar, cannula, stopper, injection port, Luer lock (all parts fit together to form a unit).
Trocar: stainless steel.
Cannula: PTFE (Poly Tetra Fluoro Ethylene), FEP (Fluorinated Ethylene Propylene), PUR (Polyurethane).
Size selected: 24G (0.70 x 19mm) yellow.
Initial sterilisation method: Ethylene oxide gas.
Packaging and labelling:
Primary packaging: Unit of use.
One (1) IV cannula in an individual sterilised peel pack.
Labelling on the primary packaging:
Name and/or trademark of the manufacturer.
Manufacturer's product reference.
Type of product and main characteristics.
If the packaging is not transparent, it must bear a diagram (preferably actual size) showing the essential parts of the product and indicating the position of the product in the packaging.
The word "sterile" (or equivalent harmonised symbol).
Sterilisation method (or equivalent harmonised symbol).
Lot number prefixed by the word "LOT"(or equivalent harmonised symbol).
Expiry date by year and month, prefixed by the word "EXP"(or equivalent harmonised symbol).
The words "for single use" (or equivalent harmonised symbol).
The words "check the integrity of the individual sterilised pack before use" (if space allows).
The words "destroy after use" (if space allows).
Secondary packaging: Protected unit.
One (1) box of 50 IV cannulas.
Labelling on the secondary packaging:
Labelling to be the same as primary packaging.
Extra information required:
Number of units per secondary packaging.
Information for particular storage conditions (temperature, pressure, light, humidity, etc.), as appropriate (or equivalent harmonised symbol).
Information for handling, if applicable (or equivalent harmonised symbol).
Manufacturer's instruction for use.
Alternatively, the instruction for use can be indicated on a separate insert.
Estimated weight: 0.277kg
Estimated volume: 1.904cdm
Instructions for use:
Sterile IV cannula to be introduced into a peripheral vein for infusing fluids, administering IV drugs.
The selected sizes are the most commonly used:
- 16G (1.70 x 45mm) grey, flow rate approx. 180ml/min, for adults: IV infusion.
- 18G (1.30 X 45mm) green, flow rate approx. 80ml/min, for adults: IV infusion.
- 20G (1.00 X 32mm) pink, flow rate approx. 55ml/min, for adults and infants: IV infusion.
- 22G (0.90 X 25mm) blue, flow rate approx. 33ml/min, for infants and children: IV infusion.
- 24G (0.70 X 19mm) yellow, flow rate approx. 20ml/min, for children and neonate: IV infusion.
Conditions for stock:
Avoid storage at extreme temperatures and humidity levels.
Check the integrity of each unit before use.
Single use material, supplied in sterile packaging, may only be used if the packaging is undamaged.
IV cannula is for single use only.
Rules of asepsis must be followed when inserting IV cannula.
IV cannula should not be left in situ for more than 72 hours. It should be removed immediately in case of infection sign.
Protection of users:
To prevent stick injuries, wear gloves.
NEVER recap the trocar after use and dispose of the needle into a puncture proof safety container.
Safety boxes are available as a UNICEF standard warehouse item reference: S0782208 - Safety box f.used syrgs/ndls 5lt/BOX-25.
Collect and destroy puncture proof containers either by incinerating them in controlled surroundings or dispose of them in a safe burial pit in compliance with national laws and regulation on health care waste management.
For more information on waste management, please refer to WHO publication "Safe Management of Wastes from Health Care Activities".
Component of a kit:
S9908302 - Obstetric, surgical kit, suppl.3-renewable
S9906627 - IEHK2006, kit, suppl.3-renewable
S9906730 - Diarrhoeal Disease Set Packing
|Component of kits|
|To see icon descriptions, go to |
' Product attribute guide'