Scissors, Mayo, 140 mm, curved, blunt / blunt.
Indicative Price 1.05 USD
Scissors, Mayo, 140 mm, curved, blunt/blunt.
Surgical dissecting scissors used for cutting tissue, threads and dressings.
See Martin Catalogue no. 24 Ref: 11 171 14.
Instrument that cuts by shearing: operating, dissecting scissors.
Curved, with blunt end blades.
Length: 14 cm.
Material: Martensitic steel (quenched, magnetic steel).
Martensitic steel composition: 0.15-0.40 % carbon; 13-14 % chromium.
Storage and transport conditions: 1°C to 30°C
Humidity: 30% RH to 70% RH
Quality Management System:
ISO 13485:2016: Medical devices - Quality management systems
EU Medical Devices Directive 93/42/ECC, Class I
Compliance to safety & product performance standards:
ISO 7153-1:2016: Surgical instruments - Materials - Part 1: Metals.
ISO 7741:1986: Instruments for surgery - Scissors and shears - General requirements and test methods.
ISO 13402:1995: Surgical and dental hand instruments - Determination of resistance against autoclaving, corrosion and thermal exposure.
ISO 17664:2017: Processing of health care products - Information to be provided by the medical device manufacturer for the processing of medical devices.
ISO 14971:2007 Medical devices - Application of risk management to medical devices.
GMDN code: General-purpose surgical scissors, reusable (38727)
UMDNS code: Scissors, Dissecting (13492)
This item must be cleaned, disinfected after each use, and sterilised in a steam steriliser.
Packaging and labelling:
Primary packaging: Unit of use.
One (1) scissors in a plastic bag.
Secondary packaging: Protected unit.
Ten (10) scissors in a box.
Labelling on the primary packaging:
Name and/or trademark of the manufacturer.
Manufacturer's product reference.
Type of product and main characteristics.
Information for particular storage conditions (temperature, pressure, light, humidity, etc.), as appropriate (or equivalent harmonised symbol).
Information for handling, if applicable (or equivalent harmonised symbol).
Over packaging: Packaging unit
Labelling on the packaging unit: Labelling to be the same as primary packaging.
Extra information required: Number of units per box.
Any symbol used on the labelling should be in accordance of the ISO 15223- 1: 2016:
If the item has a CE marking (or equivalent), the CE logo (or equivalent) must be visible on the packaging.
Component of a kit:
S9910001 - Surg. inst., basic surgery /SET
S9910003 - Surg. inst., delivery /SET.
|Component of kits|
|To see icon descriptions, go to |
' Product attribute guide'