Single-use sharps containers with an integrally attached closure feature, stackable, capacity between 5 - 7 L.
Indicative Price 1.44 USD
Sharps containers constructed of material resistant to penetration and
leakage in accordance with ISO23907:2012, compact in size allowing free
space, freestanding, lid premounted, one piece with aperture and open
and close day lid, with total capacity between 7.0 to 10.0 L, portable ,
with or without handle, single-use.
Intended use: safely and efficiently contain, transport and store used
limited number of small sharps (ex. Lancets and needles) at the point of
use and until final destruction
Functionality: the sharps container must safely contain contaminated
sharps at the point of use; during temporary storage and during handling
and transport to the point of treatment and final disposal.
Container stability: Upright, stable during use on a horizontal surface
and do not topple over when tested accordance to ISO23907:2012
Container: pre-mounted can with lid
Resistance to damage or leakage: no evidence of leakage and no breach
of the sharps
containment area when tested in accordance to ISO23907:2012
Resistance to penetration: when tested in accordance to ISO23907:2012
Total Capacity: between 7.0 to 10.0 L
Fill line indicator: Preferably with a fill line indicator marking the
maximum filling volume at a level no greater than 85 % of the total
capacity of the container and determined by the design of the container
and considering the risk of sharps extending above the fill line.
Aperture: designed to minimize the potential for accidental sharps
injuries during placement of sharps into the container. Allow single
hand placing of sharps and designed to restrict hand entry and removal
of contents from the container.
Closure feature: Integrally attached capable of being closed without the
risk of sharps injury to the user. The permanent closure, once
activated, preferably, is resistant to manual opening.
Lid: premounted and one piece with aperture
Color: Container can: yellow.
Product and packaging: fully comply with ISO23907:2012
Quality management system (QMS): Products and manufacturer must be
certified and comply with ISO 9001 requirements (or equivalent).
Packaging and labelling
Primary packaging: One box
Marking or labelling on the container is visible and easily legible and
include the following information.
• A clear indication of the fill line
• Appropriate hazard symbols.
• The word "DANGER" or the equivalent wording in English
• Identification of the specific use (where applicable) of the
container (An indication that the container is not re-usable.
• Identification of the total and/or fill volume of the container.
• Name and address of the manufacturer
• Name and/or registered trade-mark of the manufacturer including
product code or model number
• Lot or batch identification.
Protective packaging of a sturdy export quality, and of a commercial
standard that will provide adequate protection of the goods for carriage
by air, sea and/or road to final destinations worldwide, including
remote locations under adverse climatic and storage conditions, and high
humidity - i.e. not less than 17kN edge crush resistance with minimum
60% remaining with 90% humidity at a temperature of 40C (tropical
Labelling on the secondary packaging
• Name and address of the manufacturer.
• Manufacturer's product reference.
• Name and/or registered trade-mark of the manufacturer
• Description of the contents, including Type of product and main
• The number of units per secondary packaging
• Gross Weight;
• Cubic Measurement;
• Batch Number Reference (if applicable);
• Manufacturing Date (if applicable);
• Manufacturer's instruction for use given either on the package or
on a separate insert
Instructions for use
Instructions for use, required for the safe assembly, use, closure ,
handling and storage of the container, shall be made available by the
The instructions for use should minimize the potential for accidental
sharps injury. Drawings, pictograms or other graphical aids may be used
Instructions for use shall include the following, as applicable:
• instructions for proper and secure assembly of the container
before usecorrect method for placement of the sharp in the container;
• correct filling of the container to the fill line, including
specific instructions not to overfill;
• correct, verifiable closure of the container when the contents
have reached the fill line;
• correct procedure for lifting or handling the container when it
has been filled (to fill line only) and closed as per manufacturer’s
• any other warnings or precautions that the manufacturer deems
appropriate to assist the user in the safe use of the container.
S0782311 Syringe,RUP 2ml w/o ndl
S0782312 Syringe,RUP 5ml w/o ndl
S0782313 Syringe,RUP 10ml w/o ndl
S0002600 Incinerator,medical infect. waste&access
|Component of kits|
|To see icon descriptions, go to |
' Product attribute guide'