Flow meter, 0 – 5 LPM oxygen
Medical gas flow meter (Thorpe tube design) to regulate and measure the flow rate of medical gas supplied to a patient, maximum flow rate 5 LPM, for medical oxygen. Minimum order quantity (MOQ): 150 pieces OPTIONAL: ISO 32: colour coding for medical gases (indicate when ordering).
Indicative Price 27.59 USD
Medical gas flow meter (Thorpe tube design) to regulate and measure theflow rate of medical gas supplied to a patient, maximum flow rate 5 LPM,for medical oxygen.
Minimum order quantity (MOQ): 150 pieces
OPTIONAL: ISO 32: colour coding for medical gases (indicate when ordering).
Thorpe tube design flow meter using DISS input connector for panel /regulator mounting.
Can be cleaned with damp cloth and a disinfection solution.
Transparent, graduated (metric system) column, shatter resistant polymer certified for medical use.
Graduation shall have over 270 degrees of visibility.
Needle valve and body constructed of brass or aluminium.
Calibrated at 50-55 psi / 345-380 kPa / 3.4-3.8 bar inlet gauge pressure.
Inlet pressure (nominal) > 50 psi (345 kPa / 3.45 bar), peak inlet pressure at least 100 psi (690 kPa / 6.89 bar).
Pressure compensated design to give specified accuracy for whole range of input pressures
Inlet and outlet connectors DISS style, other types of connectors available upon request
Adjustment knobs to have rough surface to prevent slipping.
0 - 5 L/min, accuracy 10%, tube gradation max 0.5 L/min
1 x Outlet connector for 6 mm barbed connection to oxygen tubing included.
1 x User manual including instructions for assembly and maintenance in English, French and Spanish language.
1 x Certificate of calibration and inspection
ADDITIONAL ACCESSORIES AND EQUIPMENT, NOT SUPPLIED:
Additional adapters for inlet and outlet, if required, can be specified by the purchaser
Operating conditions: 5 to 45 deg. C, 15 to 90% relative humidity, non-condensing
Storage conditions (packaged): -40 to 60 deg. C, 15 to 95% relative humidity, non-condensing
ESTIMATED LEAD TIME:
Weight: 0.213 kg
Volume: 0.8 dm3
COMPONENTS OF A KIT
REGULATION & CONFORMITY REQUIREMENTS
CE mark conforming to Medical Device Directive 93/42/EEC
USA FDA clearance
SAFETY & PRODUCT STANDARDS
Complies with the following standards:
ISO 13485:2003 Medical devices -Quality management systems -- Requirements for regulatory purposes
ISO 15002: Flow-metering devices for connection to terminal units of medical gas pipeline systems
ISO 32: colour coding for medical gases.
Unit presentation: minimum order quantity (MOQ): 150 pieces
Labelling on primary packaging must include:
- Name and/or trademark of the manufacturer
- Manufacturer's product reference
- Type of product and main characteristics
- If the packaging is not transparent, it must bear a diagram showing the essential parts of the product
- Information for particular storage conditions (temperature, pressure, light, humidity, etc.), as appropriate (or equivalent harmonised symbol)
- Information for handling, if applicable (or equivalent harmonised symbol)
Over packaging: Packaging unit
Labelling on the packaging unit: Labelling to be the same as primary packaging
Number of units per box: 75 pieces
|To see icon descriptions, go to |
' Product attribute guide'